1、The machine positioning accuracy, reliable performance, can be adapted to the general radiography. The use of bridge rectifier combined X-ray generator, can improve the quality of radiography.
2、Adopt grid to ensure the image is more clear, especially for the post-animal radiography.
3、Configuration filament regulator circuit, the parameters more stable.
4、Designed for animals radiography, the operation is more convenient and simple.
二、The main technical parameters:
1、 Power supply conditions and operating mode
220V±22V； Power supply voltage: AC 220V ± 22V;
≥8KVA； Power capacity: ≥ 8KVA
１Ω Power supply allows maximum internal resistance: 1Ω
Power supply allows maximum internal resistance: 1Ω
2、The highest rated capacity is shown in Table 1
Tube current（mA）Tube voltage（KV）Time（s）
15 90 6.3
30 90 6.3
60 90 4.0
100 80 3.2
3、radiography conditions Tube voltage: 50-90kV
Tube current: 15,30,60,100 mA total of 4 shifts
Time: 0.08s-6.3 s a total of 19 shifts, according to R10 'coefficient selected.
4、Maximum output power
( 80kV 100mA 0.1S ) 5.92KVA.
5、Nominal electric power:
( 90kV 60mA 0.1S ) 4.00KVA.
6、Input power: 5.92KVA.
When the X-ray generator window is down, the distance between the focus and the film is 1000mm.
The X-ray generator has a rotation angle of ± 90º around the vertical axis;