1、The machine positioning accuracy, reliable performance, which can be adapted to the general radiography. The use of bridge rectifier combined X-ray generator that can improve the quality of radiography.
2、Adopt grid to ensure the image is more clear, especially for the post-animal raidography.
3、Designed for animals radiography, the operation is more convenient and simple.
二、The main technical parameters：
Power supply conditions and operating mode
Power supply voltage: AC 220V ± 22V;
Power frequency: 50Hz ± 0.5Hz;
Power capacity: ≥ 5KV
Power supply allows maximum internal resistance: 1Ω
Operation mode: intermittent loading continuous operation
2、 The highest rated capacity is shown in Table 1
Table 1. Maximum rated capacity
3 radiography conditions：
Tube voltage: 40-90kV a total of 11 shifts, each shift 5kV;
Tube current: 30,50mA total 2 shifts;
Time: 0.1s-6.0 s a total of 19 shifts, according to R10 'coefficient selected.
4 、 Maximum output power
80kV 50mA 0.1S 2.96KVA.
5、 Nominal electric power
90kV 30mA 0.1S 1.998KVA.
6、 input power：2.96KVA.
7、 Mechanical performance
When the x-ray generator window is down, the distance between the focus and the film is 1000mm;
X-ray generator around the horizontal axis angle of ± 90º;
The X-ray generator has a rotation angle of ± 90º around the vertical axis;
8、 Overall dimensions